RigExpert AA-55 ZOOM Option Bluetooth. How it works?
The structure diagram of RigExpert AA-55 ZOOM Option Bluetooth is located below:
The “brain” of the analyzer is STM 32-bit RISC microcontroller. It controls a Si5351 PLL chip running at 500 MHz and generating two sinusoidal signals. Both signals are low-pass filtered and then put through pulse shapers, producing two square signals with CMOS levels. Resistive brigde was chosen to measure parameters of a load because of its simplicity and good frequency response. The switch commutates two outputs of the bridge. After the switch, the signal is mixed with the second channel output to produce audio frequency of 1.5 kHz. This signal is then filtered and fed on the 12-bit ADC of the microcontroller. The 320×240 color TFT display and the 6×3 keys keypad are connected directly to the CPU. The USB interface allows connecting the analyzer to the PC.
Close look at the bridge
This is a diagram of the resistive bridge and its connection to the mixer: